General Description. The LP and LP are micropower voltage regulators with very low dropout voltage (typically 40mV at light loads and mV at. LPLPRE3 Texas Instruments LDO Voltage Regulators FIXED MICROPWR VLTG REG datasheet, inventory, & pricing. The LP and LP are micropower voltage regulators with very Careful design of the LP/LP has minimized all . See MIL/AERO Datasheet.
|Published (Last):||24 November 2006|
|PDF File Size:||5.24 Mb|
|ePub File Size:||15.27 Mb|
|Price:||Free* [*Free Regsitration Required]|
LP Datasheet UTC pdf data sheet FREE from
Both have a thin and very delicate metalized membrane for measurement of alpha radiation. I suspect that no long term damage should result from this misuse, as long as one does not exceed other component ratings.
If I connect the 4. The counter input INT0 is datazheet with a green arrow.
I had to jump! The rainbow dahasheet leading off screen goes to the two pushbuttons and the LCD. I just tried taking a regulator part not on a board and applied from 4. Well, I connected a 4. I do that sort of thing all the time, and I have had no problems.
It looks like the detector is on the same side as the display – does this mean you are checking yourself for radiation rather than pointing it at a source in front of you? It features an internal V power supply lo2950 mA total draw! I can always use diodes to switch between solar and backup supply, but I thought I could float charge the battery without such trickery.
We would probably buy a few, especially if the footprint was compact. Shop Blog Forum Support.
There is one error in the published circuit: I do intend to buy a few more — the sale price is very attractive, so be sure and save me a couple! Thanks eatasheet checking on the regulator — maybe I shorted the power supply while hooking things up. The JTAG fuse comes set, which makes some pins start up set high.
Not realizing this we designed our main board which lp29950 the pushbutton power switch with one of these connected to the OFF pin. Can you try doing the regulator test again? Llp2950, I disconnected it quickly enough to prevent any obvious permanent damage to the regulator. My most recent project with the ATtiny26 board is a Geiger counter. What I like about the ATtiny26 board is the convenient arrangement of two ports with access to power l2p950 ground at several points, coupled with the well-laid out prototyping area.
I expect that the regulator can tolerate this but it might add some unnecessary current drain. There is a small current draw increase at 5. The LP data sheet does not give any insight into this situation. By the way, there are a few manufacturers of the part, and your board probably has the one from On Semiconductor.
Did you look at the datasheet from that company?
Datasheets search archive of electronic components datasheets
The program currently occupies out of bytes of program memory. See below for the exterior appearance. The HV generator is top right, leading to the connector for the V tube bias red arrow. What puzzles me is that the regulator encounters this situation every time the external source is switched off and there are capacitors in the circuit that must be discharged, so there is some reason to think that such a transient backflow is not damaging.
Can you tell me what you like about it? By the way, I really like the board design. Is the RS option useful? Dwtasheet C program and the two buttons provide a menu-based system to set up data collection parameters and do the counting.
Where did you get your Geiger tube?
ATtiny26 board / LP regulator – Electronics – Pololu Forum
If only the ATtiny26 had a bit more memory… Thanks, Jim. I could post the entire project if you think it would help sales of the ATtiny26 board I do intend to buy a few more — the sale price is very attractive, so be sure and save me a couple!
The board is shown below.
We even had some ATmega8 versions made, but l2950 never sold them there were some other factors besides the ATtiny26 board not doing well, such as the mega coming out and the move to 6-pin ISP connectors. Of course we had these boards made in a batch, so now each time I populate one I have to bypass the power switch with tweezers to to clear that fuse. My power supply current-limited at about 1 amp.
In my version, the tube is screened so that small sources can be set right on top of the detector. Beta and gamma radiation are much more penetrating so the direction the tube is pointing is less important and is actually irrelevant for gamma radiation.
I could post the entire project if you think it would help sales of the ATtiny26 board. Since there is some interest, my Geiger counter is based on a circuit described by Tom Napier in Circuit CellarNovember The unit is currently monitoring the cosmic background radiation in my office 95 counts in seconds.
I especially like the plated through holes, which are fantastic for prototyping if you have to remount components several times. For one low-power project, I have the possibility of powering the board from either a 9V solar cell or a 4.